Wind speed and direction data:
Wind speed and direction are calculated every 10 minutes and sampled at 1 Hz. The trend is the average value of the vector and the factor is the highest value of 3 seconds in the sampling period. The sensor is mounted on a boom 3.2 meters above sea level.
Temperature and humidity data:
ENC produced in accordance with the issuance of standards number 57 issued by IHO.
Atmospheric pressure data:
The air pressure is the average value of one minute calculated from the sensor installed inside the buoy.
The amount of rainfall:
The sensor is mounted on a buoy mast approximately 2 meters above sea level
Radiation and Irradiation:
Kipp & Zonen sensors are proposed, the pyrgeometer (long wave) and pyranometer (short wave) sensors.
Temperature and conductivity of water:
The series salinity and temperature will carry 7 CTDs,
the Seabird SBE 37 (4 units)
and the Seabird SBE16 + (3 units)
SBE16 + is used to integrate with PAR sensor. There is one tool in a buoy well, and six sensors are installed on the monitoring cable at a depth of 10, 20, 30, 50, 75 and 100 meters and data is collected from it during maintenance operations.
Marine Currents Data:
Teledyne / RDI ADCP 300kHz is used. Measurements can be provided at a depth of about 150 meters. The device is located 6-8 m below the float in a cage, and is part of the monitoring cable.
The wave sensor is integrated with the data recorder. Directional waves are sampled at 1 Hz every hour, providing a time series of 1024 data points to be analyzed in terms of time and frequency. Many variables are derived describing the state of the sea and transmitted instantaneously to the receiving station.
pCO2 from Sami is a reagent based colorimetry sensor and is self-contained for up to one year of operation. The range is 150-700ppm with the capacity of approximately 9000 measurements with respect to the battery consumption. For one year of operation this will be equivalent of 1 per hour.
A measure of the acidity of the water PH:
Through the Seafet Satlantic sensor.
This sensor operates in the range of 2--12 pH with an internal accuracy of 0.01. It is installed in one of the wells in the buoy housing and connected to the data logger via cable.
The Wetlab FLNTUS sensor measures turbidity. The sampling capacity is 90,000 samples with a sampling rate of 8 Hz.
SBE 63 from Seabird is proposed, clamped to the mooring. The sensor is also available combined in the SBE37ODO sensor. The sensor is logging internally.
Light radiation PAR (Photosynthetic Available Radiation):
With the Satlantic PAR sensor, it measures the number of photons in the range of 400-700 nm. The sensor is located in one of the cavities in the float and is connected to a sensor. The sensor is connected to the SBE16 + sensor for recording.